|Minimum Order Quantity||1 Unit|
|Outlet Pressure||20 psi/138 kPa|
|Power||AC 220V+-10%,50 Hz|
|Country of Origin||Made in India|
|Oxygen Concentration Purity||>94%|
|Net Weight||14.3 Kg|
|Noise Level||<40 dBA|
|Size||610 x 300 x 305 mm|
|Oxygen Delivery Pressure||0.07 Mpa|
|Power Consumption||240 W|
LPM-203 Oxygen Concentrator is one of the most efficient, quietest, lightweight, high purity of 5 liter oxygen concentrator in the market today. It is designed with all the best quality Medical grade components.typically use pressure swing adsorption (PSA) technology and are used very widely for oxygen provision in healthcare applications, especially where liquid or pressurized oxygen is too dangerous or inconvenient, such as in homes or in portable clinics.
Life Plus Oxygen concentrators LPM-203 are also used to provide an economical source of oxygen in industrial processes, where they are also known as oxygen gas generators or oxygen generation plants. Oxygen concentrators utilize a molecular sieve to adsorb gases and operate on the principle of rapid pressure swing adsorption of atmospheric nitrogen onto zeolite minerals and then venting the nitrogen. This type of adsorption system is therefore functionally a nitrogen scrubber leaving the other atmospheric gases to pass through. This leaves oxygen as the primary gas remaining. PSA technology is a reliable and economical technique for small to mid-scale oxygen generation, with cryogenic separation more suitable at higher volumes and external delivery generally more suitable for small volumes.
At high pressure, the porous zeolite adsorbs large quantities of nitrogen, due to its large surface area and chemical character. After the oxygen and other free components are collected the pressure drops which allows nitrogen to desorb.
Life Plus Oxygen concentrators LPM-203 has an air compressor, two cylinders filled with zeolite pellets, a pressure equalizing reservoir, and some valves and tubes. In the first half-cycle the first cylinder receives air from the compressor, which lasts about 3 seconds. During that time the pressure in the first cylinder rises from atmospheric to about 1.5 times normal atmospheric pressure (typically 20 psi/138 kPa gauge, or 2.36 atmospheres absolute) and the zeolite becomes saturated with nitrogen. As the first cylinder reaches near pure oxygen (there are small amounts of argon, CO2, water vapour, radon and other minor atmospheric components) in the first half-cycle, a valve opens and the oxygen enriched gas flows to the pressure equalizing reservoir, which connects to the patients oxygen hose. At the end of the first half of the cycle, there is another valve position change so that the air from the compressor is directed to the 2nd cylinder. Pressure in the first cylinder drops as the enriched oxygen moves into the reservoir, allowing the nitrogen to be desorbed back into gas. Part way through the second half of the cycle there is another valve position change to vent the gas in the first cylinder back into the ambient atmosphere, keeping the concentration of oxygen in the pressure equalizing reservoir from falling below about 90%. The pressure in the hose delivering oxygen from the equalizing reservoir is kept steady by a pressure reducing valve.
Low cost of oxygen production
No other raw material needed except Air.
High purity of oxygen
Adjustable flow meter. 0-5LPM
Secure and reliable operation
Low Power Consumption
Low noise operation.
High / Low /Normal Pressure Alarm & Indicator
Handle for Easy Carrying.
Reviewed by 31 Users
Most Relevant Reviews